Are you familiar with this type of pressure sensor?
I cannot remember when is the last time where I calibrated this type of pressure instrument that is why I am excited to present it with you.
I was amazed the first time I saw it, because it is so small but has the capability to sense and measure pressure.
We also have a hard time locating it because of its small size and the way it is being installed, but luckily, we found it deep inside the testing machine.
I rarely see this type of pressure sensor and maybe it is a good thing to share it with you in case you are not familiar with it.
I will share in this post the setup and procedure on how we perform the calibration including a simple verification using the manufacturer specs as a tolerance.
Visit my other post to learn more about calibration and verification.
What is a pressure sensor?
A pressure sensor which is also known as pressure transducer is a device that is powered electronically and senses a pressure to be converted to an electrical signal.
This conversion to a electical signal allows it to be used and monitored by a programmable controllers and other similar electronic devices.
A pressure sensor of this type ( Make – Model: SMC – PSE 548-R04) are used in a special way. It is a leak tester installed in a machine that is used to test a car battery housing. It detects the leak through a pressure variation over time.
This is a pneumatic pressure sensor, it means that it is using air as the fluid. And therefore, during calibration, we will also use a pneumatic pump to generate pressure.
How to calibrate a pressure sensor?
Most pressure sensors or transducers can be calibrated by taking its electrical signal as the ouput, usually a voltage or a current then calculates its equivalent pressure values.
Read more below for the calibration setup and procedure
What to use to calibrate pressure sensors?
We will use a pneumatic hand pump with a pressure module connected to a process calibrator to simulate pressure.
Pressure Sensor Calibration Setup
Pressure sensor calibration setup- showing the input ports for supplying pressure
1. This is an SMC PSE 548-R04 type pressure sensor, so the input port is using only a quick connect tube.
2. Connect in the port as shown above the output of the pneumatic pump.
3. If you have the exact connector, just directly plug it in the port. You are now ready to supply the necessary pressure.
Pressure Sensor Calibration procedure.
1. Once it is properly setup and has stabilized, pump the necessary pressure.
2. Supply an increasing pressure. S
3. Then apply
4. Record in your Measurement Data Sheet. Perform 2 trials
How to P
We will be using the manufacturer specs as a basis for tolerance to determine a pass or fail results.
- The pressure sensor has an accuracy of 2% of FS
- Fulls Scale reading is = 1 Mpa
tolerance = 0.02 x 1 = .02 Mpa
- Therefore, reading should be within +/- 0.02 Mpa
I have presented here, for me, is a rarely seen pressure sensor where I believe others also is not familiar about it. I also discussed a simple calibration and verification procedure using a pneumatic pump with a pressure module.
Thank you for visiting my site.
Please comment and subscribe.
You can also connect with me in my Facebook page.