Temperature and humidity are the most monitored environmental factors based on the facilities I have visited. You can see it almost everywhere.
Have you ever experienced being exposed to a high temperature and humidity environment? I believe all have experienced it, but not just aware regarding the effects of temperature and humidity in our daily activities or some of our activities.
In this post, I will present below topics:
1. Importance of monitoring these environmental conditions, specifically temperature and humidity in a calibration laboratory.
2. The negative effects of having them very high or very low as applicable to some process or activities.
3.The related clause and requirements as per ISO 17025:2017
4. The requirements to observe when choosing a monitoring instruments for temperature and humidity.
I will also introduce the instruments used to measure both temperature and humidity in a calibration process.
In our Daily Activities……..
Have you ever experience
During summer where temperature and humidity is high, have you ever experienced in your skin that when you perspire, it stays longer that feels very uncomfortable? And during winter where the air is very dry that it makes our skin and throat also dry?
These feelings (discomfort and irritation) we encountered are the effects of temperature and humidity in our body. In our home, wooden types of furniture that gets very dry tends to crack or expand. Because of this, we should be familiar with the importance of temperature and humidity monitoring to prevent and control these things that happen in our daily processes.
Ideally, Relative Humidity (RH) should be within the range of 30% to 50% and the temperature is within 20 to 25 deg Celsius. Above this will cause fungus or molds to grow, and below this will cause dryness and discomfort.
Some common examples of why we need to monitor and control temperature and humidity are:
To prevent the formation of molds and fungus that can cause diseases,
- Maintaining a certain product fresh in a storage room where fruits and vegetables are stored,
- To monitor the environment for fermentation like in beer or wine making or for incubation of eggs,
- To ensure maintenance of good quality in some special personal collections like paintings, guitar or clothing,
- To give a healthy environment, for example, maintaining a good air quality to avoid dryness on skin and eyes that causes discomfort and breathing related illness like asthma.
Temperature and Humidity in Calibration Laboratory
As required in the ISO 17025:2017, the requirements under clause 6.3, the laboratory environmental conditions should be monitored and controlled. Two of this environmental conditions are the temperature and humidity.
Monitoring of environmental conditions is a requirement under ISO 17025:2017. Below are the clauses:
- Clause 6.3 Facilities and environmental conditions, with sub clauses:
- 6.3.1 The facilities and environmental conditions shall be suitable for the laboratory activities and shall not adversely affect the validity of results.
- 6.3.2 The requirements for facilities and environmental conditions necessary for the performance of the laboratory activities shall be documented.
- 6.3.3 The laboratory shall monitor, control and record environmental conditions in accordance with relevant specifications, methods or procedures or where they influence the validity of the results.
Environmental conditions that are controlled means that there is a specified range or conditions of acceptability of that certain parameter like temperature and humidity.
It should be monitored, means that there is a specified interval for checking if it is meeting the acceptable range/conditions or not, and it should be recorded for the purpose of reviewing the results and as an evidence of monitoring and documentation.
Furthermore, monitoring and control are not just during the calibration process but also during storage and handling of equipment which also includes the calibration status of measuring instrument (example, the thermo-hygrometer).
This is how important it is to consider these two parameters because failing to follow and consider them can cause a lot of problems where discontinue of calibration work is imposed if this requirement of humidity and temperature is not met.
And technically speaking, this is not just applicable to instruments in a laboratory but also in equipment or machines used in manufacturing that is why it is also controlled inside the facilities of every manufacturing company.
Temperature And Humidity in Calibration Reports
Moreover, temperature and humidity is not only recorded in a the monitoring sheet, it is also required to be recorded in the calibration or test report during the actual execution of the laboratory activity like during calibration. Below is the related clause where it is required to be recorded on the calibration report.
- Clause 7.8.4 Specific requirements for calibration certificates
- Calibration certificate shall include b) the conditions (e.g. environmental) under which the calibrations were made that have an influence on the measurement results;
>> Calibration certificates, test reports or simply measurement reports should reflect the environmental conditions at the time of performing the measurement. This is to ensure that we perform the calibration on the acceptable environmental conditions where it has no effect on the validity of the results and enable us to repeat the process when necessary.
Does Temperature Affect Humidity?
Temperature can affect the humidity, they are inversely proportional, meaning, as temperature increases, humidity decreases, and vise versa. So in order to control humidity, we need also to control temperature.
What are the Effects of Temperature and Humidity in our Instruments and Calibration Process
There are guides and recommended practice for the use of temperature and humidity in a laboratory. You may find it through NIST handbook, ISO guides, or NCSL on their websites.
Requirements for a dimensional lab for temperature and relative humidity (RH) is 20 +/-1 with an RH of 30% to 40%
Electrical or electronics is within 23 +/-3 Deg C temperature and a humidity of 35% to 55%.
Please take note that provided requirements is not fixated on every lab, it is still based on the equipment specifications so the best thing to do is to check first all instruments specs before performing the calibration.
It is a best practice to record all the necessary environmental requirements specifically temperature and humidity of a specific instrument for calibration for easy reference.
Reasons Why we Monitor and Control Temperature and Humidity Inside a Calibration Laboratory
While monitoring temperature and humidity is a requirement of the Standards, below are some of the main reasons:
- In pressure calibration, density changes with temperature.
- Dimensional calibration where the use of gauge blocks and other dimensional instruments are affected by thermal change usually a temperature higher than 20 Deg C. If you are calibrating or measuring highly accurate instruments, this will be a problem.
- With a humidity >60%, metals like gauge blocks will rust quickly.
- In an electrical area, when metals rust or corrosion occurs, this will create a leakage path for voltage and current which causes a malfunction or inaccurate reading.
- If below 20%, where the air is very dry, it will be prone to static electricity which can damage electronic equipment.
- Most of the manufactured instruments have a requirement on its environmental condition in order to function on its full accuracy. You may check it on its specifications.
Because of these reasons or effect of uncontrolled parameters, temperature and humidity are included as one important factor in the area of quality control. This is also to ensure that any detected errors in the readings of calibrated instruments are not the cause of temperature and humidity variations.
The Requirements for Instruments Used to Measure and Monitor Temperature and Humidity
There are no specific requirements or specifications of a thermometer or hygrometer that the standard requires to be used during monitoring, but it is advisable to use a continuous recorder of temperature and humidity that is available and directly seen or observed inside the laboratory.
If you are searching for an instrument to be used for monitoring temperature and humidity, below are some criteria/requirements to meet (as per my experienced during audits):
- The temperature and humidity monitoring instrument should be calibrated before and during use;
- The temperature and humidity output display must be easily accessed during the performance of calibration – to be recorded in the data sheet in real time;
- Previous data can be extracted for review or if requested;
- Recorded data can be stored for a defined period.
In monitoring a temperature and humidity, we can use a separate thermometer and a humidity meter or hygrometer. But why waste our resources if we can have both in one instrument. There is now the Thermo-hygrometer, a hygrometer thermometer meter in one. Today’s advancement, a chart recorder is also included depending on the manufacturer. The advantage of monitoring and recording are already in place..
For the procedure in calibrating a Thermo-hygrometer and other details, please visit my other post in this link.
Extech Humidity-Temperature Chart Recorder Review
Temperature and humidity (relative humidity) monitoring involves data recording and review. It is where we based our analysis about their effect whether it is acceptable or not. Since there is analysis and review involve, we need to have a clear picture of what is to be analyzed.
This is where this instrument are a big help, it does not only display the reading but it also records and display data in a chart from a previous hours or days. Monitoring will be easy because you can directly observe the highest or lowest reading based on its chart.
In the lab that I worked for,Extech RH520A Humidity and Temperature Chart Recorder (Model RH520) is the Thermo-hygrometer that we are using. Its capability meets our expectations where it is a perfect tool for the job (as per my experience).
What I Like in this product (Pros/Features)
- Its probe (where sensors are located) can be mounted to different position depending on your needs. It has a stretchable cable ( 1 meter) to be positioned easily. You can place the probe inside a confined space if you need to without including the whole unit like inside a refrigerator.
- The unit can be used with a battery ( AA) or power supply and therefore can be portable where it can be carried to a different remote location. It also designed to be mounted on a wall or just placed it on a table top.
- It has a memory where data is stored and can be extracted on a defined schedule.
- Programming or set up is easy to follow basing it on its user manual, user-friendly.
- Temperature and humidity are digitally displayed plus a chart to view the past and current state. Also to view and analyze if temperature difference or fluctuations has happened. Display are large enough for you to see it easily in a distance.
- Data can be extracted to excel format in a computer ( Laptop) to further review and analyze ( need to install the software provided).
- It can also display date and time where monitoring can be analyzed together with temperature and humidity.
- With a low battery indicator
- Alarm can be set for proper timing and notifications when needed.
What I Dislike in this product (Cons)
- I am not sure if this is including all the models but sometimes when I connect it to my laptop, it has a difficulty in communicating. It will search first for the exact port before communication or connection takes place.
- The storage memory is not that very big where data should be downloaded on a defined schedule (say 1 month, but I guess this is already good enough if download schedule is shorter), then reset it again to clear and have a new fresh start to save and record a new data. This is applicable if you need the past data for storage and future review.
- Needs to reset time and date if the power is accidentally turned off (battery can be a back-up power if installed) or battery is changed (not a big deal).
Who its for
This is primarily useful almost in every temperature and humidity monitoring, especially in a room environment monitoring. If you want to make your monitoring more detailed with some automated features, this one is for you.
|weight||357g / 12.6oz|
|temperature Range||-28°C to 60°C(-20 to 140°F )|
|Temperature Accuracy||±1°C / ±1.8°F|
|Humidity Range||10 to 95%|
|Humidity Accuracy||±3.0% typical|
|Internal Memory Storage||49 152 complete reading sets|
|Sample Interval||selectable recording rate: 0.1 to 199.9 minutes|
|Graph scaling||User selectable in 10 , 5 and 10% increment|
For Extech RH520A Calibration Procedure, please visit this link
Temperature and humidity are the 2 most monitored environmental conditions in a laboratory where neglecting to monitor, control, and maintain these will result to a non-conformances during an audit.
In this article, I have presented below topics:
1. Importance of monitoring these environmental conditions, specifically temperature and humidity,
2. The negative effects of having them very high or very low as applicable to some process or activities.
3. The related clauses and requirements as per ISO 17025:2017.
4. The requirements that we need to observe when choosing the instrument used to monitor temperature and humidity
Thank you for visiting my site. Please leave a comment and subscribe.
You can also connect with me on my Facebook page.